nbsp; Classification of adsorption isotherms and brief analysis of adsorption mechanism nbsp; nbsp;The adsorption isotherm of BR is the source of information about the pore structure, adsorption heat and other physicochemical characteristics of adsorbents. Under the condition of constant temperature and wide range of relative pressure, the adsorption isotherm of the adsorbate can be obtained. In order to better understand the information contained in the adsorption isotherms, the following is a brief introduction to the classification of adsorption isotherms and adsorption mechanism [1, 8, 10, 19, 20, 33, 53 ~ 55]: many adsorption isotherms can be divided into six types (IUPAC classification), as shown in the figure is the type of adsorption isotherms. For microporous adsorbents with a very small surface area, the adsorption performance is type I adsorption isotherm. The type I adsorption isotherm and the partial pressure P/PO line are concave and high-frequency experiments are characterized by the formation of a platform, which is horizontal or nearly horizontal. As the saturation pressure reaches, the adsorption isotherm either directly intersects with P/po=1 or shows a "tail". The initial part of the adsorption isotherm represents the filling process of narrow micropores in the adsorbent, and its limit adsorption capacity depends on the accessible micropore volume rather than the surface area. The slope of the platform under high relative pressure is caused by multilayer adsorption on non microporous surfaces (such as mesopores or macropores and outer surfaces). The normal type II adsorption isotherm is a loose single-layer to multi-layer adsorption caused by nonporous or macroporous adsorbents. The existence of inflection point indicates the transition from monolayer adsorption to multilayer adsorption, that is, the completion of monolayer adsorption and the beginning of multilayer adsorption. The type III adsorption isotherm is usually related to the weak adsorbent adsorbate interaction and the strong adsorbent adsorbate interaction. In this case, the synergistic effect leads to the formation of multi-layer adsorption before the completion of a uniform single adsorption layer, so the adsorption capacity increases rapidly with the progress of adsorption, The interaction between adsorbate and adsorbate plays an important role in the adsorption process. Water vapor adsorption on non porous surfaces is the best example of type III adsorption isotherms. The obvious feature of type IV adsorption isotherm is that it has a hysteresis loop, which is closely related to the occurrence of capillary condensation, and maintains a constant adsorption capacity in a high and wide partial pressure range. Its initial part is similar to type II adsorption isotherm, which corresponds to single-layer to multi-layer adsorption on the mesopore wall. Some mesoporous or microporous carbons in few adsorbents show V-type water "br"
adsorption isotherms. Like type III adsorption isotherms, the interaction between adsorbent and adsorbate is very weak compared with that between adsorbent and adsorbate, which of course includes the formation of hydrogen bonds by water molecules. Type VI adsorption isotherms are quite rare, but they have special theoretical significance. They represent multilayer adsorption gradually formed on homogeneous non porous surfaces such as graphitized carbon, and each step height provides the adsorption capacity of different adsorption layers. Br
Fig, The adsorption isotherm n (P) can be expressed by the following formula [56 ~ 60]: "br"
n (P) is the number of adsorption moles when the relative pressure is p; Br
Hmin and Hmax are the smallest and largest pore sizes of adsorbents; 《br》
ρ (P, H) is the molar density of n when the aperture is h and the relative pressure is p; Br
f (H) is the pore volume function corresponding to the pore width h, which characterizes the pore size distribution of the adsorbent. "Br"
Evans  and his colleagues studied the performance of the liquid adsorbed in the pores by using the approximate method of statistical thermodynamics, namely the mean field density function theory. The mean field density function theory is an approximate theory that divides the interaction between liquid molecules in the mean liquid phase into short-range repulsion and long-range gravitation. The effect of long-range interaction on liquid properties can be treated approximately according to the theory of mean field density function; The short-range repulsion effect is modeled by the equivalent arrangement of hard spheres. The prediction of the adsorption equilibrium phase in the larger pores by the mean field density function theory is equivalent to the adsorption model proposed based on the thermodynamic analysis method, while the description of the adsorbed molecules in the smaller pores is closer to the real situation. In particular, it can predict the thickness of the adsorption layer on the pore wall, and can show the change from capillary condensation to pore filling under the critical pore size. The interaction between molecules is calculated by Lennard Jones potential energy equation as described above. In 1989, seaton and others first used the density function theory to measure the pore size distribution of carbon adsorbent from micropore to mesopore. With br
, Oliver and conklin (micromeriticsins.corp.) proposed a more general calculation method. They used the regularization method of nonlocal or uniform density approximation (nonlocalorsmootheddensityapprox is 28 pin dual in-line package image) to further expand to the large hole range (0.4nm to 400nm). Using DFT method, the author  and some foreign scholars [59 ~ 62] characterized the pore size distribution of carbonaceous adsorbents. The results show that DFT is a simple algorithm based on high-resolution adsorption isotherm, and it can characterize the pore range (micropore to macropore) of the whole porous solid with only one method. In addition, the current general molecular simulation method is Monte Carlo method [63 ~ 65], which is similar to the aforementioned DFT method and will not be repeated here. In a word, from the current research of carbon adsorbents and the papers published in relevant publications, the molecular simulation and pore size distribution of carbon adsorbents are the current research hotspots. Br
Company Profile br
bester Instrument Technology (Beijing) Co., Ltd. (huihaihong nanotechnology) is a famous manufacturer of specific surface area testers in China. It was the first company specializing in the R & D, production, sales, maintenance, technical support, training and after-sales service of full-automatic nitrogen adsorption specific surface area testers. It is a high-tech enterprise recognized by Beijing Zhongguancun Science and technology park. Br
the full-automatic nitrogen adsorption specific surface area tester developed and produced by our company is widely used in the R & D, production, analysis and monitoring of graphite, battery, rare earth, ceramics, alumina, chemical industry and University powder materials. After years of efforts, our company has a large number of customers. The advanced technology, stable performance and good after-sales service of instruments make our universal experimental machine products enjoy a high reputation and trust among customers. Br
3h-2000iii specific surface instrument is a famous brand in China, with a large number of customer groups (see the company's station for details). It is the latest model with high test accuracy and good repeatability, and it is the only model in China with a repeatability of 1.5%. 3h-2000 series specific surface tester is the most widely used model in China, which is widely used in the testing of battery graphite, rubber carbon black, University, catalysis and other powder materials, including coinage, customs, Chinese Academy of Sciences and other units br
3h-2000 Ⅲ specific surface area tester performance characteristics br
1, high measurement accuracy, good reproducibility, and the relative error of repeatability is less than ± 1.5% (the only one in China); Br
2. The measurement range is wide, and the specific surface area can be measured at 0 × Materials within the range of 103m2/g meet the measurement of specific surface area of all powder materials and porous materials; Br
3. The test speed is fast, with an average of about 5 minutes for each sample, which is only one quarter of the test time of similar foreign instruments; Br
4. The low-temperature nitrogen adsorption dynamic chromatography is adopted, and the unique parallel gas circuit at home and abroad eliminates the external effects such as ambient temperature, humidity, atmospheric pressure, etc., without vacuum pumping, and the laboratory conditions are relatively loose; Br
5. The instrument integrated degassing device (non split, domestic exclusive) solves the problem of degassing and testing integration, realizes the in-situ treatment of samples, only needs one installation, and has zero contact with air, ensuring the efficiency and effectiveness of sample pretreatment; Br
6. Double 40 liter steel cylinders, double gas sources, can be used normally for more than 2 years; Br
7. The software is a self-developed high-precision data processing system, 16 bit data acquisition card, with a minimum resolution of ＜ 0.01mv, fully automatic trigger, ensuring high measurement accuracy and sensitivity, stable operation, and compatible with Win98, winme, WIN2000, WINXP and other operating systems, USB interface; Br
8. The instrument software has excellent operation stability and reliability, excellent performance, and the hardware adopts stable and durable key components, which is suitable for frequent testing; Br
9. The instrument is easy to operate and learn, with complete accessories, considerate after-sales service, a wide range of customers and high reputation. The earliest model of "br"
3h-2000 series instrument is hpdai-88 high-efficiency dynamic adsorption instrument, which is gb/t (original gb/t) according to the national standard. This model passed the identification of the science and Technology Bureau of the Ministry of chemical industry on August 5, 1988, and the technical identification certificate number is:  chemical identification No. 48. After improvement and upgrading, the 3h-2000 series instrument passed the national standard inp4h on December 16, 2002. The water contact angle of Pu hydrogel prepared by UV curing is 30 (4) 0b, which is the same as other hydrogels used to make contact with eyes and shows good anti protein adhesion. The certificate number is: jjzz No. 2002- (17) -017. Br
BET specific surface area meter specific surface, specific surface area, specific surface area meter, specific surface area meter, specific surface tester, specific surface area tester, specific surface tester, specific surface tester, specific surface analyzer, specific surface analyzer br
BET specific surface area meter specific surface, specific surface area, specific surface tester, specific surface tester, specific surface tester, specific surface tester, Specific surface analyzer《
Copyright © 2011 JIN SHI